The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||19 January 2012|
|PDF File Size:||11.70 Mb|
|ePub File Size:||6.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
Click Open to start the calibration measurement. The printing date changes when a new edition is printed. If you do any operation other than the following steps, the system may not be recovered. If anything is missing, contact the nearest Agilent Technologies office. Dangerous voltage levels, capable of causing death, are present in this instrument.
Do not replace components with the power cable connected.
Turn the key to the right until it is locked. When the recovery is finished, push the enter select OK. The EC restarts automatically. CN or below This section describes how to mabual the contents of the C drive to the factory state.
Network setting GPIB setting Printer setting The driver for the supported printer installed after purchase is deleted. E5071f dialog box disappears automatically after a few seconds. It’s a community-based project which helps to repair anything. The system recovery procedure is different in accordance with the HDD revision.
CN or above Step 1. Inset the key in the lock and turn anti-clockwise. All rights are reserved. Test and Measurement Call Center tel 1 fax 1 Canada: Click Short to start the calibration measurement.
Safety Symbols Instruction Manual symbol: Click Done to terminate the full 2-port calibration process. Press the button under the screw. The EC restart automatically. The touch screen calibration screen Figure appears. For more information, see Modifying Calibration Kit Definition. Click Port x-y Thru to start the calibration measurement x and y denote the test ports between which the THRU connection is being made.
EC User Manual – EC Network Analyzer from Keysight
To use this function, you must create the user backup image in advance. If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered. The firmware revision you installed remains even if you perform factory recovery. A total of twelve error terms, six each in the forward direction and the reverse direction, are used in a calibration. Repeat the above procedure for port y.
The following screen appears during backup. Click Port x Open to start the calibration measurement x denotes the test port to which the standard is connected.
Remove the two stickers item 1. This calibration makes it possible to perform measurements with the highest possible accuracy.
Replacement Procedure Step 1. In EC, right click My Computer. Keep Away from Live Circuits – Operators must not remove instrument covers. The volume label for C: The procedure consists of the following items.
Shut down the EC. Select the C drive to restore, then click Next button. CN or below Returns the contents of the C and D drives to a user-specified state.